Local Run Manager TruSight Oncology 500 Analysis Module does not require a sample sheet because you enter sample and run parameter information into the software interface.
For analysis on the TruSight Oncology 500 Local App, see the sample sheet for your version and follow directions in the appropriate app guide:
For RNA libraries, follow instructions in the TruSight Tumor 170 Local App User Guide with the sample sheet template TruSight Tumor 170 Local App Sample Sheet. Remove any references to DNA from the sample sheet.
TruSight Oncology 500 includes 523 genes. Of these genes, the panel covers all exons for 522.
The kit enables the preparation of 48 libraries from DNA, RNA, or a combination of DNA and RNA libraries. DNA and RNA libraries can be prepared simultaneously.
See the denature and dilute libraries guide for your sequencing
For a list of compatible sequencing systems, see the TruSight Oncology 500 compatibility page.
Covaris E220evolution, LE220, or ME220.
Customers can use their own pipeline. However, we do not provide resources to facilitate this.
Approximately 16 hours.
No, the algorithm settings cannot be changed.
Review the system requirements in the TruSight Oncology 500 Local App User Guide. When requirements are met, contact Illumina Technical Support to request the link to download the Docker or Singularity image.
No. Because of software constraints, run them separately.
Use SandyBridge, an equivelant, or better. Older processors do not have required data compression instruction sets and might cause analysis to fail.
For a list of compatible analysis software, see the TruSight Oncology 500 compatibility page.
Bioinformatic analyses for the following: